|Approvals/Requirements Satisfied by eRADIMAGING Courses|
|~ ASRT approval for ARRT® Category A credit||~ MDCB approval by the Medical Dosimetrist Certification (Selected Courses)|
|~ ARMRIT accepted (All MRI Courses)||~ CAMRT and Sonography Canada recognize the ASRT approval (All Courses)|
|~ ARDMS accepted (All Courses)||~ Florida approval for all courses 1 credit or more|
|~ NMTCB accepted (All Courses)||~ California CE requirements met for all radiography courses|
|~ All Courses eligible of international radiographers' CPD requirements||~ All Courses available for RRAs|
|~ ASRT and MDCB are approved continuing education providers of ARRT® and all courses are accepted by ARRT®|
The Benefits of Digital Radiography
Mark P. Bowes, PhD
*Medical Writer, Portland, Oregon.
Address correspondence to: Mark P. Bowes, PhD, Medical Writer, 7135 SE 18th Avenue, Portland, OR, 97202. E-mail: firstname.lastname@example.org.
Disclosure Statement: Dr Bowes reports having no financial or advisory relationships with corporate organizations related to this activity.
Since the introduction of digital radiography 3 decades ago, digital systems have largely replaced screen-film radiography at many imaging facilities in the United States. Digital radiography systems use a variety of specialized X-ray detectors, conversion methods, and processing algorithms to convert X-ray energy to numeric data for electronic display, transmission, and storage. In general, digital systems may be categorized as either computed radiography or digital radiography. Computed radiography uses a cassette-based system that is similar to conventional screen-film cassettes, with a separate laser scanning process to extract X-ray intensity data. In contrast, digital radiography uses several approaches to directly convert X-ray energy to digital data without the need for a separate scanning step. Digital detectors are able to produce clinically useful images across a much broader range of input radiation doses than film-based imaging. As a result, it can be difficult for the technologist to determine when a patient is receiving more radiation than needed to obtain a useable image. Exposure indicators and deviation index values have been developed to provide the technologist with feedback about the intensity of radiation received at the detector, and about how this intensity compares with a preset target value. As with screen-film radiography, successful digital imaging requires consideration of many technical factors, including collimation, exposure control, and the use of antiscatter grids. Digital radiography may provide several advantages in comparison with screen-film radiography, including greater ability to manipulate image processing and acquisition parameters, compatibility with electronic recording-keeping and archiving systems, reduced need for film and film processing materials, smaller storage space, and shorter imaging times. This article provides an overview of potential advantages and limitations of digital methods of X-ray detection and processing, how to identify appropriate radiation exposure levels, and other technical considerations.
Full Course Content available to active members of eRADIMAGING.com
* This sample course is for reference purposes only. It is not currently available for earning CE credits. To earn ARRT CE credits please subscribe to eRADIMAGING where you will see a complete listing of all active and eligible CE courses.